On March 26 there was a brief power outage and the power the shared 700 wasn't restored before the UPS quit. There was a fuse in the third floor panel room had tripped and that impacted the power for just this unit and the 800 magnet pump. After resetting the fuse power was restored and the cryoprobe warm-up was completed. The turbo pump was kept running overnight to achieve cold box vacuum and on March 27 cool-down is underway.
700 - shared cryoprobe power is recycled - power failure